Autoclave Steriliser Vacuum Steam Fully Automatic 23L Class N with Printer for Dental Clinics Beauty Nail Tattoo Salon
1. 4 new technology
Multi-protection safety lock
Wide liquid crystal screen
Steam compensation techniques
Doublestage tandem type vacuum pump
2. Japan vacuum pump
Noise is less than 60db, vacuum pump can highly reach -0.8KG/cm2, vacuum pumping efficiency is 30% higher than similar sterilizer.
3. Italian CEME liquid pump
4. Italian CEME valve body
5. PARKER valve body
6. Three times pulsation vacuum, the final vacuum can be up to -0.85bar, which ensure the air can be evacuated efficiently and achieve to saturated-vapor state. Thus the steam can completely permeate into the article with lace fabrics, hollow appliances (like handpiece) and packing articles, which is very important to sterilizing.
7. The humidity after Vacuum drying is less than 0.2%. The pulsing work equipped with double series vacuum pump when in the process of whole pulsing.
8. Separate steam generation technology, appliances are safer, disinfection are faster. Combines the steam compensation technique, ensure the 134 degreen moist heat sterlization as required by appliance.
9. Preset 8 concrete programs are listed according to their actual frequency of use. And what's more, added a hemophilus inactivation program.
10. Automatic door open and close system.
11. Fully automatic operation, the whole process computer self check and monitor.
12. Disintection process self evaluation system.
1. LCD Display, 2 program easy to choose
2. Automatically feeding water and excess water and air to be drained away.
3. Show the reason of failure on LCD directly
4. Class N model, sterilize unpackage solid instrument, used widely in beauty and tatoo salon.
5. Safety Devices:
a. Over heat control: When inner temperature is over heated, power will be shut down automatically and alarm system
b. Over pressure control: It have alarm system when over tempeature
c. Safety door lock system
Chamber Dimensions(mm)(Φ×D): 250×490
Chamber Capacity(liter): 23L
Overall Size(cm)(L×W×H): 48×69×42
Power supply: 230VAC ± 10% 50HZ/10A
Control type: Full automatic microprocessor
Bacteria filter: Standard
Appliance tray: Stand 3 tayers
Operation enviroment: +2-40 indoor operation
Max relative humidity: 85%
Sterilizer Construction: 121/134
Stainless steel instrument container: Stainless steel SUS#304
Care and Maintenance:
Disinfect and clean the inside of the sterilizer with medical alcohol every month.
Change the fuse, when the fuse is bad, you must turn off the power first, and then change the fuse.
Clean the seal termly: You should clean the seal termly, to prevent the impurity, and improve sealability. If the sealability is bad, you should debus the seal and clean it. If the seal is bad, you must change it in time.
Change the seal:
a) When the seal is invalidation, you should change it in time. You may nip the border of the seal and take off it.
b)Clean the trough of fixing seal.
C). Kindly press the new seal into the trough of fixing seal till it enter completely. Don't use the knife-edge to stick or press the seal, such as screwdrever, knife and so on.
Transportation and storage:
1) The preparative of transportation and storage: Turn off the power, pull out the power plug, and cool the sterilizer.
2) The condition of transportation and storage: Circumstance temperature:-20℃-+55℃, relative humidity:<85%